Stone disease in the children is becoming more commonplace. Over the past 25 years the incidence has increased approximately 6-10% annually and is now 50 per 100,000 adolescents 1. The diagnosis of kidney stones in children, as in adults, relies primarily on diagnostic imaging. In adults, the most common imaging study performed in the United States for the initial diagnosis of kidney stones is a computed tomography (CT) scan, due to its superior sensitivity and specificity. This is not preferred for children as CT utilizes ionizing radiation and children have an increased sensitivity to radiation effects. In addition, stone formers often have recurrent stone episodes over their lifetime, which is especially relevant to younger stone formers. The repeated exposures of a CT scan could lead to an increased risks of secondary cancers2. As a result, ultrasound is often performed in children instead because there is no radiation associated with its use1. Ultrasound, however, is less sensitive and specific compared with CT, and is known to overestimate kidney stone size3-5, which is one of the primary determinants of how stones are managed.
We have identified a new technique to improve the accuracy of US stone sizing. Traditionally, a kidney stone will show up as a bright, or hyperechoic, object on US, and radiologists measure the longest linear dimension of the bright area to represent the stone size. We believe that the dark, or hypoechoic, acoustic shadow that appears behind a kidney stone provides additional information for predicting stone size. The resolution of the stone is affected by the distortion of the waves traveling through the intervening tissues; the resolution of the shadow is only affected by the local stone obstruction.
We screened 660 stone diagnoses over an 11 year period (2004 – 2014) at Seattle Children’s Hospital, a tertiary care referral center serving the greater Pacific Northwest. Over the study period, there were 37 patients presenting with an initial diagnosis of a kidney stone, and who had both a US and CT within three months of each other. Two reviewers retrospectively measured both the stone size and the shadow width from the ultrasound image. We compared the results to the stone size measured from the CT scan. A total of 48 stones were included in the study with an average size based on CT imaging of 7.85 mm. The shadow width was present in 88% of the cases, and, on average, was more accurate than measuring the stone itself. Measuring the stone width on ultrasound tended to overestimate the size of the stone by 1.2 ± 2.5 mm (reviewer 1) and 2.0 ± 1.7 mm (reviewer 2), while measuring the shadow width on ultrasound underestimated the size of the stone by -0.6 ± 2.5 mm (reviewer 1) and overestimated the stone size by 0.3 ± 1.2 mm (reviewer 2). In both cases, the shadow was a better predictor of stone size and, for reviewer 2, there was less variability in the data. Stone sizes based on CT are typically considered within 1 mm, with low variability.
Our technique is simple and can be easily adopted by pediatricians, radiologists, and urologists. It improves the accuracy of US and gives physicians more confidence that the reported size is more representative of the true stone size, without having to expose children to the radiation of a CT scan. This also allows the physician to make more accurate decisions about when to perform a surgery for a large stone or continue to observe a small stone that may pass on its own. The findings from this study may also potentially be applicable to stones in adult patients. We believe that with continued advancements in US, we can reduce the number of CT scans for both adults and children. The results in part highlight one of the drawbacks of ultrasound, which is user dependence; two users can have very different results. It is anticipated that future work on automated stone and shadow sizing can reduce this along with standardizing the method in which the shadow measurement is taken.
1. Tasian G and Copelovitch L. Evaluation and Management of Kidney Stones in Children. J Urol. 2014: Epub ahead of print
2. Pearce MS, Salotti JA, Little MP, et al. Radiation exposure from CT scans in childhood and subsequent risk of leukaemia and brain tumours: a retrospective cohort study. Lancet 2012; 380: 499-505.
3. Ray AA, Ghiculete D, Pace KT, Honey RJD. Limitations to ultrasound in the detection and measurement of urinary tract calculi. Urology 2010; 76(2):295-300.
4. Fowler KAB, Locken JA, Duchesne JH, Williamson MR. US for detecting renal calculi with nonehnanced CT as a reference standard. Radiology 2002; 222(1):109-113.
5. Dunmire B, Lee FC, Hsi RS, Cunitz BW, Paun M, Bailey MR, Sorensen MD, Harper JD. Tools to improve the accuracy of kidney stone sizing with ultrasound. Aug 2014; [Epub ahead of print].
Franklin C. Lee1 – email@example.com
Jonathan D. Harper 1 – firstname.lastname@example.org
Thomas S. Lendvay1, 2 – Thomas.email@example.com
Ziyue Liu3 – firstname.lastname@example.org
Barbrina Dunmire 4 – email@example.com
Manjiri Dighe 5 – firstname.lastname@example.org
Michael Bailey – email@example.com
Mathew D. Sorensen1, 6 – firstname.lastname@example.org
University of Washington
1 Department of Urology, Box 356510
5 Department of Radiology, Box 357115
1959 NE Pacific St, Seattle, WA 98195
2 Seattle Children’s Hospital
Urology, Developmental Pediatrics
4800 Sand Point WA NE, Seattle, WA 98105
3 Indiana University
Department of Biostatistics
410 W. Tenth St, Suite 3000, Indianapolis, IN 46202
4 University of Washington
Applied Physics Lab – Center for Industrial and Medical Ultrasound2
1013 NE 40th St, Seattle, WA 98105
6 Department of Veteran Affairs Medical Center
Division of Urology
1660 South Columbian Way, Seattle, WA 98108